How to install the machine correctly? The machine parameters may be incorrect during transportation or operation
Power supply conditions:
Please configure the correct power supply according to the name plate of the machine
The error range of input voltage should be within ± 10%, and the machine should be correctly grounded to prevent damage caused by electric leakage.
1. There is no strong vibration source and strong electromagnetic field around
2. It should be placed in a stable and horizontal position without serious dust and direct sunlight
3. Machine installation space: at least 50cm at the back, 30cm at the side and 100cm at the front
4. Operating environment requirements: room temperature (ambient temperature 5 ~ 40 ℃, relative humidity no more than 85%)
How to adjust the machine parameters that may be incorrect during transportation or operation?
Please call us at + 86 13794832878
08:00 - 20:00, Monday To Friday
GCI Testing Instrument Co., Ltd
Tel：+86 0769-22322732 Mobile：13794832878(wechat)
Address：No. 41, East Zone 2, Xinxing Street, Datangtou, Dongcheng District, Dongguan City
Copyright © GCI All Rights Reserved.
Pay attention to wechat
Browse mobile site